Home › Forums › Mooshimeter Support › Can't measure 120v AC › Reply To: Can't measure 120v AC
December 5, 2018 at 12:15 pm #19992
You have samples set to manual 32. Auto offers 256.
If i assume (60 Hz?) and calculate right, your settings let mooshimeter see about half of the sine wave you are measuring.
If you set samples to automatic (or just 256) there would be almost 4 full waves to analyze per round. Should give better results then.