Home › Forums › Mooshimeter Support › meter stopped working after update › Reply To: meter stopped working after update
February 7, 2015 at 7:50 pm #2137
When the firmware on the meter is updated sometimes the phone can be confused because the Mooshimeter will be broadcasting with the same address but other connection parameters will have changed. If you haven’t already, can you try:
- Turning Bluetooth off and on again?
- Rebooting your phone?
Also – just checking – is the Mooshimeter blinking? Sometimes if a firmware update goes wrong the meter will revert in to the bootloader and blink the LED on and off at a rate of once per second. If the meter is giving one very faint blink every 10 seconds, the firmware update was OK.