Home › Forums › Mooshimeter Support › crashing after update
- This topic has 6 replies, 2 voices, and was last updated 8 years, 2 months ago by admin.
- adam reedGuest
After installing latest update the app crashes if you turn on xy. It also randomly crashes when trying to connect to the meter but after an attempt or two will then connect. Im using android on a galaxy tab 4.
Thanks Adam… now that crash reports are centralized I can take a look at the logs. Will try to see where it’s coming from.
I found out it doesn’t crash on start up as long as no other blue tooth device is connected.
Very interesting… I haven’t run that experiment before. Can you tell me what other device you’re using? Funny enough I don’t actually have any other bluetooth devices to test with…
Milwaukee m12 speaker http://www.milwaukeetool.com/power-tools/cordless/2592-20
I’ve also noticed the app always crashes when I select xy mode in the graph. I have 2 android devices both running KitKat, a Nexus 7 (2012) and Samsung galaxy 5 mini. Both suffer from the same bug.
Thanks Adam, I’ll run an experiment.
Thanks for directing me to this. I found the bug and uploaded a new version. You should be able to use XY mode again after an update.
It was actually a version control issue that caused the production build to use an old version of the graphing library.