- This topic has 1 reply, 2 voices, and was last updated 4 years, 4 months ago by Anonymous.
- May 10, 2018 at 10:18 am #19790farzadGuest
We are dealing with small currents and reading the value in amp on the app is so inconvenient and error prone. How can we change the unit from amp to micro amp?
also what is the minimum amp value mooshi can measure?
- May 10, 2018 at 10:43 am #19791AnonymousInactive
There is only one range for current measurement, which goes up to 10 amps with about 5 microamp resolution. Some milli amps would be more realistic, as test leads pick up noise and couple last digits are basically random anyhow.
I would suggest using external shunt resistor for extremely small currents, but it would mean that you need to do way more mental conversions on the fly, this time from (milli)volts and probably with stranger ratio.
And again, free math channel would help, but unfortunately it is not (yet?) implemented.