- This topic has 1 reply, 1 voice, and was last updated 7 years, 6 months ago by Anonymous.
- May 21, 2016 at 7:04 am #9803Pim VeldGuest
Just received my Mooshimeter (V.1 FW Build: 1445847981) . I use an iPad mini (model MD532SL/A) under iOS 9.3.1 with it. I tested it by measuring the voltage of a wall socket (240 Vac). The meter starts up in Vdc. When I change it over to Vac the measuring frequency increases (which I think is normal) and after a very short time (1 second to sometimes a few seconds) the meter terminates the measurements, the App moves to the scanning window showing the mooshimeter. I have to tap the line with the mooshimeter again, a small ‘delay-circle’ appears at the right side of the line and the whole process start over again.
At it is now this meter is not suitable to carry out measurements.
Another problem is that in Graph mode the App closes down when the screen rotates (with the iPad).
- May 21, 2016 at 7:50 am #9804AnonymousGuest
I red some more forum-posts and have updated the firmware to Build 1447470000. This solved both problems. Actually the firmware-update via iOS did not proceed like the YouTube video for iOS. I had to press the reset-button first, like in the video for Android. Subsequent Bluetooth connection happend automatically (or maybe I touched something inadvertently?)
Is it correct that graphs cannot be shown in landscape orientation?