Home › Forums › Mooshimeter Support › no connection
- This topic has 4 replies, 2 voices, and was last updated 7 years, 5 months ago by admin.
- Knud NollinGuest
I can not connect to my Mooshimeter. At first it was only my ipad who could not connected to my Mooshimeter,, but now it is also my iphon who can not connect .. I’ve tried to reset, remove the batteries, connected ohms and common. But nothing helps. The LED is flashing 2 short and 2 long flashes all the time. My iphon can see Mooshimeter but not connect.
If the LED is giving long flashes continuously that probably indicates firmware corruption. Does the app try to upload firmware when you connect to it?
What version of the app are you using? You should be able to tell by touching the gear icon in the top right of the “scan” screen. If the gear icon does nothing or is not present, it means you are using a very old version.
There is a new version of the iOS app and firmware in the Apple approval pipeline – if you want to try it out earlier I can add you to the beta group but I need your permission and email address.
Mooshimeter iOS App Version: 1.03(71) my iPhone i up to date with. iOS 8.2. The led is blinking 2 short and 2 long and has Done so for the last 24 hours. It continius even after removal of the battery, and renset.
My e mail is firstname.lastname@example.org Feel frie to use it….
Best.. Knud Nollin
I am also getting the 2 long, 2 quick orange flashes…
Is there a fix?
Hi Wondering Geo,
Your meter might be in hibernation mode – is it brand new or have you used it before?
If it’s in hibernation, you can bring it out of hibernation by connecting the “C” and “Ω” inputs together and holding them together for 10 seconds. The meter will reboot (you will see the orange flashes again) and then the Mooshimeter’s radio will be on. There should have been an instruction card that came with your meter.
Let me know how it goes, best