- This topic has 0 replies, 1 voice, and was last updated 6 years, 1 month ago by Anonymous.
- June 30, 2016 at 5:22 am #10451AnonymousGuest
It looks like the iPhone beta app might be having trouble parsing the data that comes back from the meter when the main voltage input is set to AC. When I hook my meter up to a 120v outlet and set the input to AC mode it goes from showing a normal, expected ~0.0v value (which bounces around a little due to noise) to showing OUT OF RANGE. This seems to show up for all sample rates/buffer sizes, with one exception; at 125hz I see values around 30-35v. Any sample rate higher than that just returns OUT OF RANGE.
What’s interesting is that if I set the meter up to log while I’m doing this, when the app is showing OOR the logfile shows ~120v values, so it looks like the meter itself is measuring correctly and that the problem is actually on the app side of things.
Loving the new version, btw!
- June 30, 2016 at 4:37 pm #10457adminKeymaster
I found a pretty embarrassing ranging bug in the iOS beta release 1.07(836). I just put up a new version 1.07(946) that should address it.
Hope this fixes it, let me know how it goes
- July 1, 2016 at 8:58 pm #10478AnonymousGuest
Looks good now. You rock!