- This topic has 0 replies, 1 voice, and was last updated 4 years, 8 months ago by Anonymous.
- September 8, 2017 at 12:09 pm #18850AnonymousInactive
Hey Mooshim Team,
My mooshimeter was logging throughout last night (current + voltage while charging some NiMH batteries in an LED toy) – and after viewing the log I noticed it’s complete useless. There occurred an failure code 460 after ~6hours and after that all ampere values are really strange, like12345678910[...]1504828171.210, 4.441729E-02, 6.674085E0001504828173.465, 4.445095E-02, 6.675294E000CH1: AMPS 10ACH2: VOLT 60VSMPL: 0256 0125HzUTC TIME SEC ,CH1 AMPS DC ,CH2 VOLT DC1504828209.221,-1.766873E025, 6.676045E000,0004601504828211.692,-1.766873E025, 6.673865E000[...]
Complete Log (1,3MB): https://drive.google.com/open?id=0BxEZsMPKTcshWVlKVkswcXBaLVE
(I changed the Sampling depth from 256 to 64 at beginning – for better reading on app and then back to 256 to reach a 2 second log interval – interestingly this generated, opposite as I expected, no log entry)
negative ampere? I don’t think so ^^ looks like an maximum negative value… (the batteries was charged fine btw)
I found here, that “Code 460 means the meter has just rebooted and is resuming logging”, but there must happen something else.
I have another log with failure 460 (there I changed nothing while logging, it just appeared at some point, but continued logging fine after that.123456781504355772.684, 6.960216E-02, 5.128813E0001504355775.817, 6.957331E-02, 5.128831E000CH1: AMPS 10ACH2: VOLT 60VSMPL: 0256 0125HzUTC TIME SEC ,CH1 AMPS DC ,CH2 VOLT DC1504355777.349, 6.957331E-02, 5.129033E000,0004601504355779.606, 6.957812E-02, 5.128996E000
So what happened? Any Idea? Maybe this helps bugfixing…
Firmware is 1477971088