- This topic has 1 reply, 2 voices, and was last updated 4 years, 10 months ago by Anonymous.
- November 6, 2018 at 2:51 am #19948vivienGuest
this subject has already been discussed, however not for the 3 last years.
And now, with the current version of the mooshimeter app on iOs, I can see there is a choice for Thermocouple type K however is anyone understanding how it works ? When I plug the probe to my thermocouple type K it does not work : out of range
- November 6, 2018 at 8:32 am #19949AnonymousInactive
I get “out of range” when i have open contact somewhere. Does your thermocouple work with other devices? Did you connect it right? It should go between common and auxiliary voltage (ohm-connector).
What do you get for temperature, if you short common and ohm, while app tries to measure thermocouple? Aux voltage should go to zero and internal and thermocouple temperatures should be nearly same.
Thermocouple need to be connected in right polarity, otherwise temperature readings start to change in wrong direction. Pinching your probe between fingers should give instant change towards your body temperature (assuming air temperature around you differs from it. Very easy to test here in Finland ;) ).